TY - JOUR

T1 - A Transverse Oscillation Approach for Estimation of Three-Dimensional Velocity Vectors, Part II: Experimental Validation

AU - Pihl, Michael Johannes

AU - Stuart, Matthias Bo

AU - Tomov, Borislav Gueorguiev

AU - Rasmussen, Morten Fischer

AU - Jensen, Jørgen Arendt

PY - 2014

Y1 - 2014

N2 - The 3-D transverse oscillation method is investigated by estimating 3-D velocities in an experimental flowrigsystem. Measurements of the synthesized transverse oscillatingfields are presented as well. The method employs a 2-D transducer; decouples the velocity estimation; and estimates the axial, transverse, and elevation velocity components simultaneously. Data are acquired using a research ultrasound scanner. The velocity measurements are conducted with steady flow in sixteen different directions. For a specific flow direction with [α,β]= [45,15]°, the mean estimated velocity vector at the center of the vessel is (vx,vy,vz) = (33.8,34.5,15.2) ± (4.6,5.0,0.6)cm/s where the expected velocity is (34.2,34.2,13.0)cm/s. The velocity magnitude is 50.6 ± 5.2cm/s with a bias of 0.7cm/s.The flow angles α and β are estimated as 45.6 ± 4.9° and 17.6± 1.0°. Subsequently, the precision and accuracy are calculated over the entire velocity profiles. On average for all direction,the relative mean bias of the velocity magnitude is −0.08%. For α and β, the mean bias is −0.2° and −1.5°. The relative standard deviations of the velocity magnitude ranges from 8 to 16%. For the flow angles, the ranges of the mean angular deviations are 5° to 16° and 0.7° and 8°.

AB - The 3-D transverse oscillation method is investigated by estimating 3-D velocities in an experimental flowrigsystem. Measurements of the synthesized transverse oscillatingfields are presented as well. The method employs a 2-D transducer; decouples the velocity estimation; and estimates the axial, transverse, and elevation velocity components simultaneously. Data are acquired using a research ultrasound scanner. The velocity measurements are conducted with steady flow in sixteen different directions. For a specific flow direction with [α,β]= [45,15]°, the mean estimated velocity vector at the center of the vessel is (vx,vy,vz) = (33.8,34.5,15.2) ± (4.6,5.0,0.6)cm/s where the expected velocity is (34.2,34.2,13.0)cm/s. The velocity magnitude is 50.6 ± 5.2cm/s with a bias of 0.7cm/s.The flow angles α and β are estimated as 45.6 ± 4.9° and 17.6± 1.0°. Subsequently, the precision and accuracy are calculated over the entire velocity profiles. On average for all direction,the relative mean bias of the velocity magnitude is −0.08%. For α and β, the mean bias is −0.2° and −1.5°. The relative standard deviations of the velocity magnitude ranges from 8 to 16%. For the flow angles, the ranges of the mean angular deviations are 5° to 16° and 0.7° and 8°.

U2 - 10.1109/TUFF c .2013.006238

DO - 10.1109/TUFF c .2013.006238

M3 - Journal article

VL - 61

SP - 1608

EP - 1618

JO - I E E E Transactions on Ultrasonics, Ferroelectrics and Frequency Control

JF - I E E E Transactions on Ultrasonics, Ferroelectrics and Frequency Control

SN - 0885-3010

IS - 10

ER -