Publication: Research - peer-review › Journal article – Annual report year: 1965
A servosystem that controls the sample temperature in an electron spin resonance spectrometer is described. It is based upon the regulation of the combination of two nitrogen gas flows of different temperatures. The temperature can be preset with an accuracy to about 1 degC between -140 and 100°C and maintained constant to within ± 0.1 degC.
|Citations||Web of Science® Times Cited: 1|